Before first use every day
SoftShells™ should be sanitized once a day. For both Inverted Nipple and Sore Nipple Kits.
If you have just used the SoftShells, the foam inserts must be cleaned separately before sanitizing. Squeeze the milk out of the foam, then rinse them twice with clean cold water. Dry them by squeezing between a clean cloth or paper towel. Separate and place the silicone back and dome in a small pan filled with cold water. Distilled water is recommended. Bring the water to a boil and allow it to continue boiling for approximately 10 minutes. Drain off the water and place parts on a clean cloth to dry.
For Quicker Cleaning
Medela offers Quick Clean™ Micro‑Steam™ bags for daily kit disinfection and Quick Clean wipes for cleaning before and after each pumping session. Shells and backs can be put in the top rack of a dishwasher or an autoclave. Do not place foam inserts in a dishwasher, Quick Clean Micro-Steam or autoclave.
Between uses during the day
If you are leaking, special care must be taken to avoid the build-up of dried breastmilk on the shell parts and in the foam inserts. If the foam inserts become saturated, separate the shells, backs and inserts. The shells and backs should be washed with a mild soap or detergent in warm water, rinsed thoroughly in cold water and dried. Squeeze the milk out of the foam, then rinse them twice with clean cold water. Dry them by squeezing between a clean cloth or paper towel.